For parts or repair- This item is being sold as is, as pictured. As this is for parts or repair no returns will be accepted.
- Unit turns on, but we were unable to move the microtome unit with the control, just turn on both lights.
- Top section of the microtome (section that moves the microscope) is partially loose on the base.
- Microscope and controls look in good condition.
- Cryosectioning system not fully tested, just powered on.
- See video for more details.